Printing Calibrator Barcodes on the Atellica® Solution Job Aid Version 1.25 and Below
This job aid provides instructions on how to print calibrator barcodes on the Atellica® Solution.
Atellica® Solution SIEMENS Calibration Healthineers Print Calibrator Barcodes – Calibrator Definitions screen SIEMENS Healthineers Atellica Solution Version 188.8.131.5224001 1 SP00107 Calibrator barcodes must be printed using the Atellica System Samples Inventory 2 Calibration Qc Patient Orders Worklist Maintenance Setup Solution software to ensure that each barcode has the correct Calibration Overview Create Calibration Orders Calibration Results Calibrator Definitions IMT Calibration sequence number. Filters 3 Assays Material Name Lot ID Material ID Expiration Date Assays Add New CALC CC53 01/04/2019 11:59:59 PM VB12, Fer Edit To print calibrator barcodes from the Calibrator Definitions Status O CALRub G C703 12/14/2018 11:59:59 PM Deleto Expired Rub_G screen: Active TOX Cal 72909627 TO 02/28/2019 11:59:59 PM Acet, Sal 5 Print Barcode O Inactive Audit Chem 06580 AC 01/05/2019 11:59:59 PM Alb, ALT, DBil_2, Crea_2, TP, GluH_3 1. On the Command Bar, select Calibration Audit IM Print Calibrator Barcode X 2. Select the Calibrator Definitions tab A1c_3 CAL Material Name Lot ID Material ID Expiration Date CALaHAVM 3. To limit the number of Calibrator Definitions that display, CALD CALA CA98 01/15/2019 11:59:59 PM V 4 use the filters in the Filters area of the screen V CALA CALD CD94 10/16/2019 11:59:59 PM Audit TDM TOX Cal 72909627 TO 02/28/2019 11:59:59 PM 4. Select the material name for 1 or more calibrator CALHBSII DRUG CAL definitions LSP CAL 5. Select the Print Barcode button Audit 1.14.2 ➢ The system displays all selected calibration definition ? PEP Labl barcode labels in the Print Calibrator Barcode screen 6. Select Print ➢ Barcodes print on the system barcode label printer ? 6 Print Cancel 1 Effective Date: 28-DEC-2020 | HOOD05162003154972 Unrestricted © Siemens Healthcare GmbH, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Effective Date: 28-DEC-2020 | HOOD05162003154972 Unrestricted © Siemens Healthcare GmbH, 2020