Printing QC Barcodes on the Atellica® Solution Job Aid Version 1.25 and Below
This job aid provides instructions on how to print QC barcodes on the Atellica® Solution.
Atellica® Solution SIEMENS Healthineers Printing QC Barcodes Job Aid P-P1-03 SIEMENS . QC Barcodes Healthineers Atellica® Solution Version 126.96.36.19963002 1 Samples Inventory Calibration ac Patient Orders Worklist Maintenance Setup QC barcodes must be printed using the Atellica Solution Create QC Orders Control Bracketing Qc Statistics Patient QC Flagging 2 QC Definitions QC Panels QC Scheduler software to ensure that each barcode has the correct sequence Filter. Y Status: All number. Control Level Control ID Control Lot Control Lot Expiration Usage Specimen Type Date Last Used Edit/View. Control Name URINE TOX 3 UT3 66220 08/31/2018 Primary Urine 06/05/2018 To print QC barcodes: 3 3 Copy Urine 06/05/2018 URINE TOX 2 UT2 66220 08/31/2018 Primary URINE TOX 1 UT1 66220 08/31/2018 Primary Urine 06/05/2018 Delete Primary Serum 05/29/2018 At the Command Bar, select QC Urine Chemistry2 2 42766792T 66792T 12/31/2018 Export .. 1. Urine Chemistry1 42766791T Primary 05/29/2018 667911 12/31/2018 Serum TM PLUS3 3 54633 54633 09/30/2019 Priman Serum 01/18/2018 4 Print Barcode ... 2. Select the QC Definitions tab TM PLUS2 2 54632 54632 09/30/2019 Priman Serum 01/18/2018 TM PLUS1 1 54631 54631 09/30/2019 Priman Serum 01/18/2018 Select the checkbox(es) for the Control Name(s) to print sBNZ Control 3 3 BNZ3 TOX1801M 01/31/2019 Primary Serum 07/30/2018 3. SBNZ Control 2 2 BNZ2 TOX1801M 01/31/2019 Primar Serum 07/30/2018 SBNZ Control 1 BNZ1 TOX1801M 01/31/2019 Primary Serum 07/30/2018 4. Select the Print Barcode button IA PLUS 3 40353 40353 05/31/2020 Primary Serum 07/31/2018 3 LA PLUS 2 2 40352 40352 05/31/2020 Primary Serum 07/31/2018 The barcodes will print to a local barcode printer, or to a IA PLUS 1 05/31/2020 Serum 07/31/2018 40351 40351 Primary HBSIPOS 2 POS632 6315472 06/03/2018 Priman Serum 03/29/2018 networked barcode printer shared with the system. HBsll Nog NEG631 631547 06/03/2018 Priman Serum 03/29/2018 HAVM POS Primary Serum 10/03/2017 2 HAVPOS 701038 04/13/2018 L ....... ...... QC Orders .. Import Add QC Definition ... ? PEP LabManager Tuesday, July 31, 2018 11:58:26 AM NOTE: Some QC manufacturers will provide pre-labeled QC tubes. NOTE: For the Sample Handler configuration, all tubes must be barcoded. For the Direct Load configuration, tubes may be run either barcoded or non barcoded. 1 Template Effective Date: 27-MAY-2019 | HOOD05162003010284 Effective Date: 06-APR-2020 | HOOD05162003077519 Unrestricted © Siemens Healthcare GmbH, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability may vary from country to country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers 2 Effective Date: 06-APR-2020 | HOOD05162003077519 Unrestricted © Siemens Healthcare GmbH, 2020