Removing Atellica® IM Analyzer Reagent Loader Path Obstructions Video Version 1.25 and Below
This video will explain how to remove reagent loader path obstructions on the Atellica® IM Analyzer.
This video will show the procedure that can be used to remove reagent loader path obstructions on the atellica. I am analyzer 1st put the I am analyzer into the Stop State by selecting the system button on the command bar. Next select status. Then select the atellica. I am analyzer icon. If the analyzer is not already stopped, select the pause button to allow the system to complete any testing currently in process. The system will pause and then go into stop state. Then unlock the IM back cover. And open the I am back cover. You must open the cover within 30 seconds of unlocking, otherwise the cover will relock. Ensure the reagent probes are clear of the reagent compartment shutter. If not, manually move the probes to the right over the probe rent stations while avoiding pushing on the probe tips. Open the reagent compartment shutter and ensure there are no obstructions. Manually remove any obstructions from inside the compartment and from the reagent loader path. Close the reagent compartment. Close the cover. Navigate to the reagent loader screen by first selecting inventory. And then select reagent loader. Then select the atellica. I am analyzer icon, select the resume work button and ensure all loader path obstructions are cleared and select yes to complete this procedure. Resume the item analyzer by selecting the resume button on the IM System status screen to bring the analyzer back to the ready state. This completes the procedure to remove reagent loader path obstructions on the atellica. I am analyzer.
stop state,iam analyzer,analyzer icon,resume work,back cover,analyzer,resume,cover,pause,probes,compartment,hereinafter collectively,regulatory requirements,operator manual,reagent drawer,reagent loader,ancillary reagents,reagent compartment,siemens healthineers,operator,manual,patent,reagent,overview,local,healthineers,availability,material,training,' reagents
System Please note that the learning material is for training purposes only! Overview Reagent L Reagent v Reagent Needs Supplies Overview Lot Operator Di•ønostics For the proper use Of the software, please always use the Operator Manual Or Instructions for Use (hereinafter collectively "Operator Manual") D-pp-15 System issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. VMM -03 D-PP-15 00104 Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time Of the training Reagent Drawer Reagent Loader The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Standby Standby Standby Note: Some functions shown in this material are optional and might not be part of your system. Remouing Reagent Loader Path Capabilities Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. SIEMENS . SIEMENS Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. Obstructions on the [email protected] Cornpattrrert CornpattrneN CornpanrneN CornputrreN The reproduction, transmission or distribution Of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. Loader Status All names and data or patients, parameters and configuration dependent designations are fictional and examples only. Healthineers 1M Analyzer Video All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Barcode Error Connected to PCC Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands ore the property of their respective owners. Base 11 Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Base Copyright O Siemens Healthcare GmbH 2020 Reagent Patent Prcressim Reagent v Patent Rement Water Reagent Drawer Water Global Siemens Healthineers Headquarters Frcnt Cover F rmt Cover F rmnt Cover Henkestr. 127 Cwene Reagent Loader Water prim-an' Reagents Prim-a"' Reagents Ancillary Reagents AnciUary Ancillary Reagents prim-an' Reagents 91052 Erlangen, Germany 35 Back Cover Cwene Phone: +49 913184 siemens.com/heolthineers Tray Waste 755) 57 7:57 75767 75737 Resume Work Manual Entry Clear LabManag•r