Removing Sample Containers from the Atellica Magline® Transport Manually Video Version 1.25 and Below
This video will explain how to remove sample containers from the Atellica Magline® Transport manually.
This procedure can be used to manually remove sample containers from the atellica magline transport. 1st put the Atellica magline transport into stop mode by selecting the system button on the command bar and then selecting the Status tab and then the Atellica Magline Transport Module icon on the Atellica Magline Transport Status screen. Select the stop icon. Visually inspect the atellica magline transport to locate any carriers with sample containers. Remove one or more atellica magline transport covers to access the sample containers by first locating the screws under the atellica magline cover. Then loosen the two atellica magline transport cover screws enough to allow you to slide the cover off. You do not need to completely remove the screws. Pull the atellica magline transport cover away from the Atellica magline transport and set aside. Remove any sample containers from the carrier and place in Iraq. Do not remove the carrier's from the Atellica magline transport. The Atellica magline transport carriers contains strong magnets and should only be removed by authorized service personnel. Ensure carriers are clear of the atellica magline transport diverters. Diverters are areas where the atellica magline transport splits into two directions. The system will not initialize properly if carriers are blocking the diverters. If appropriate, ensure all analyzer probes in sample handler or sample handler Connect grippers are off the atellica magline transport and in a safe position before starting the system to enable free movement of the carrier's when the atellica magline transport initializes. Then reinstall the Atellica magline transport covers by positioning each cover so the edges of the cover fit into the slots on either side of the opening. Slide the Atellica magline transport cover in until the spring clips lock the cover into place. Tighten the two screws under the atellica magline in cover. To complete the procedure, start the Atellica magline transport by selecting the resume icon on the Atellica magline Transport System status screen.
sample containers,atellica,carriers,remove,cover,system,procedure,selecting,screws,slide,carrier,place,diverters,hereinafter collectively,means substitute,regulatory requirements,operator manual,future availability,local representative,specific country,training material,operator,local,availability,material,country,training,patien,note,proper
Atellica• Solution Samples Inventory Calbabon PatieN PatieN O rckrs Patienl Orders Patient Orders Maintenance Setup Seatup Please note that the learning material is for training purposes only! operator Operator Diagnostics For the proper use Of the software, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") For the proper use of the software, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") Diagnostics issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. „ssued by Siemens Healthineers_ This material is to be used as training material only and shall by no means substitute the Operator Manuel. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software System .01 CH and hardware available at the time Of the training. and hardware available at the time Of the training The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Standby Note: Some functions shown in this material are optional and might not be part of your system. Remouing Sample Containers Diagram Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Capabilities Due to regulatory requirements, the future availability Of said Functionalities in any specific country is not guaranteed. Please contact your local Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. from the AteLLica [email protected] Atellica Wine Atellica W i n e Transport Atellica Wine Transport The reproduction, transmission or distribution Of this training or its contents is not without express written authority. Offenders will be Liable for damages. The reproduction, transmission or distribution Of this training or its contents is not without express written authority. Offenders will be liable for damages. C. p I .01 c-p1-01 AIL names and data Of patients, parameters and configuration dependent designations are fictional and examples only. All names and data of patients, parameters and configuration dependent designations are fictional and examples Only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Transport ManuaLLg SK-PI -01 c-p1-01 connected to pcc Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright O Siemens Healthcare GmbH 2020 Global Siemens Headquarters Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 913184 0 siemens.com/healthineers Tuesday, 2B, Tuesday, February 2B, 2017 102234 AM Tuesday, 2B, 2017