Replacing Cuvettes on the Atellica® IM Analyzer Video Version 1.25 and Below
This video provides instructions on how to perform the task of replacing the cuvettes on the Atellica® IM Analyzer.
This video will show you the steps for how to replace Q vets on the Atellica Immuno Assay Analyzer. The inventory icon on the command bar will display a yellow or red alert when system supplies are low. After selecting the inventory icon on the Command bar, select the Supplies overview tab. And then select the immuno assay module graphic. Locate the status of the queue vets. A yellow triangle indicates the cube at supply has dropped below the bin sensor and the cube bin is almost empty of Q vets. A red triangle indicates the supply level is out. To replace the immuno assay, kovetz open the cuvet bin lid. To open the cuvet packaging, use scissors to ensure that plastic bag fragments do not drop into the cuvet bin. That could cause jams. For the cube at, since the cube had been up to the brim in the left side of the Cuvet bin. Do not push down the queue vets buschke vets to the right side of the bin, or overfill the bin enough to prevent the cuvet lid from fully closing. Also, do not remove the plastic divider. Performing any of these can cause a cuvet jam. After loading the queue vets complete the procedure by closing the Cuvet bin.
command bar,cuvet bin,cuvet,yellow,cube,side,closing,supplies,select,inventory icon,•' solution,hereinafter collectively,patent grant,utility model,system fluids,operator manual,siemens healthineers,operator,manual,system,open,reagent,healthineers,country,material,training,siemens,version 1,show,replace
SIEMENS Atellicao•' Solution Atellica"•' Solution Atellica"•' Solution Version 1 Version 1 Version 1. .12.0.2777002 sp00107 Please note that the learning material is for training purposes only! _A Reagent Lot Cornparison Lot Cal •QC Storage Inventory Cal-QC Storage Inventory Cal-QC Needs Reagent Loader Reagent Needs For the proper use Of the software or hardware, please always use the Operator Manual or Instructions for Use sp00104 sp00107 (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis System Fluids Lat Expiration Sequence and does not necessarily reflect the latest version of the software and hardware available at the time of the training. IMAcid 1M Acid 27 d 23 h 2/63/2018 2/612018 9,'2/6/2018 The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. 1M Base 0K 27 d 23 h 9,'2/6/2018 2/63/2018 0019 00194 SIEMENS Note: Some functions shown in this material are optional and might not be part Of your system. 0K 1M Cleaner 6d23h 24d2h 9,'2/6/2018 2/61/2018 1215 1M Wash Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be 27 d 23 h 24d 16 h 16 h Healthinecrs Healthineers commercially available in your country. Due to regulatory requirements, the future availability Of said Functionalities in Replace System Fluids Replace Systern Fluids Replace System FlI uids any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. Supplies Waster Status The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. 3 Storage Locatons Avadable 3 Storage Locations Avadable Tips Cuvette Tip Waste Offenders will be liable for damages. 5 Tp Trays In Storage CUvettes Almost empty Tip Tray Waste Open Tray Waste Drawer Open Tray maste Drawer All names and data Of patients, parameters and configuration dependent designations are fictional and examples only. Waster 0K Liquid Waste All rights, including rights created by patent grant or registration Of a utility model or design, are reserved. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright @ Siemens Healthcare GmbH 2020 Wedresday, wedresday, 18, Wednesday, wedresday, 18, 2017 Lab-Manag•r Law anag•r