Replacing Sample Tips on the Atellica® IM Analyzer Video Version 1.25 and Below
This video provides instructions on how to perform the task of replacing the sample tips on the Atellica® IM Analyzer.
This video will show you the steps for how to replace sample tips on the Atellica Immuno Assay Analyzer. The inventory icon on the command bar will display a yellow or red alert when system supplies are low. After selecting the inventory icon on the Command bar, select the Supplies overview tab. Then select the immunoassay module graphic. Locate the status of the tips. A yellow triangle indicates there are less than 120 tips on board and the tips are almost empty, or that the strap was not removed from a tip tray that is loaded on the system. A red triangle indicates the supply level is out. To prevent cross contamination where new, uncontaminated gloves when handling sample tips. To replace sample tips, first open the sample tip loading lid. Remove the protective cover from the sample Tip Tray bundle. Install the sample tip trays, ensuring that the rounded notches in the trays are facing the back of the system. Twist and then pull the tab down and out to release the strap from the trays. And then pull the strap out to remove it from the bundle. Be sure to leave the top on the upper tip Tray. To complete the procedure, close the sample tip loading lid.
tip tray,command bar,yellow,trays,select,tips,strap,remove,pull,bundle,1m base,tray waste,inventory icon,supplies overview,reagent overview,reagent loader,system fluids,waste,supplies,inventory,reagent,system,status,empty,open,siemens,nventory,waster,cuvette,storage
SIEMENS SIEMENS .. AtellicaTM Solution Version 1 . 12.0.2777002 SP00107 SP00104 Healthineers Heolthineers System SarnpJes Supplies Samples Inventory I nventory Calibration QC Patient Orders Worklist Maintenance Reagent Overview Supplies Overview Reagent Needs Reagent Loader Supplies Overview Reagent Overview Reagent Needs Reagent Needs cal-QC Needs Reagent Loader Supplies Overview Reagent Lot Comparison Cal•QC Storage Inventory Cal.QC Storage Inventory cal-QC Needs System Fluids Volume Status OBS Lot Expiration Sequence Number 10 00114 1M Base 11 1040 27d2h 9.123/2018 SP00104 1M Cleaner OK 0K 6 d 22 h 9/123/2018 8/125/2018 00184 00114 001048 00104 0011 25 d 23 h 6 d 22 h 8/125/2018 2223 1M Wash 1M Base 27d2h 8/125/2018 4/125/2018 9.123/2018 000028 001048 1040 Replace System Fluids Supplies Su Waster System Status Status 5 Storage Locations Available Almost empty Cuvette and Waste OK O TIP Trays In Storage OK TIP Tray Waste Tray waste OK Cuvettes Tray Waste Open TIP Tray Waste Drawer Waster Water Liquid Waste Wednesday, October 18, 2017 LabManager