Replacing System Fluids on the Atellica® CH Analyzer with Direct Load Video Version 1.25 and Below
This video provides instructions on how to perform the task of replacing system fluids on the Atellica® CH Analyzer with Direct Load.
This video will show you the steps on how to replace the Italica Ch system fluids as appropriate using the primary display. The Ch analyzer houses the following fluids that need replacing. When empty. Ch conditioner Ch cleaner. Ch diluent. NCH wash these fluids can be replaced without pausing or stopping the analyzer. The inventory icon on the command bar will alert the user when system supplies are low. To replace the system fluids, first select inventory on the command bar. Next, select the supplies Overview tab, select the chemistry icon, determine which system fluids require replacement a yellow triangle in the software indicates the bulk system. Fluid bottle is empty and the system is using the reservoir or the lot or on board. Stability will expire in less than 48 hours. A red triangle indicates the supply level of the system fluid is empty. Or there is only enough fluid in the reservoir to process aspirated tests. To change the fluids, start by pulling out the system fluids drawer. Locate the appropriate system fluid bottle that needs replacing. Tighten the cap on the system fluid container and remove the container. Discard the old container per laboratory protocol. Next, select the replace system fluids button in the software. Scan the fluid container barcode located on the front of the bottle with the handheld barcode scanner. Ensure the software displays the correct lot and lot expiration numbers. If the correct lot and lot expiration are not showing, enter the lot and lot expiration manually into the system. Select the save button in the software. Holding the new system fluid container vertically with the septum facing down, insert the container straight down into the drawer with the label facing away from the analyzer. Confirm the container is fully seated by ensuring that the tabs on the sides of the bottle are flush with the drawer surface. To prevent a vacuum from forming inside the container. That could cause the analyzer to stop processing. Vent the container in order to vent the container. Squeeze and twist the cap counterclockwise to pass the first designated stop, then no longer squeeze, but twist the cap until it reaches the next designated stop. Complete the procedure by closing the system fluid drawer.
Atellicae Solution Version 126.96.36.19922006 Atellicae Solution Atellica• Solution Version 188.8.131.5222006 AtelIicaØ Solution Version 1. SIEMENS SIEMENS , AtellicaØ Solution Version 1 Atellica$ Solution Version 184.108.40.20622006 Emulation Mode ATELLlCA-sy ATELLICA.SF ATELLICA-SY ATELLICA-SF Version 1. 16.03122006 16.0.3122006 ATELLICA.SY Healthineers• Healthineers • Healthineers C&dion Orders Inventory System C alüatbn Pdient Pdknt Invertory C all&atbn Workbt Reagent Cartridge Entry' Reagent Cartridge Entry Reagent Overview Reagent Needs Reagent Loader Supplies Overview Reagent Loader Reagent Needer Reagent Needer Reagent Loader Supplies Supplies Overview Cal-QC Storage Inventory Cal-QC Needs R eageot Need S Reagent Needs Reagent Lot Comparison Cat-QC Needs Needs Status Operator Diagnostics (Legacy CC) SH OCH SIEMENS System System Fluids System Fluids Volume Status OBS Expiration Lot Expiration Lot CH Conditioner Sequence Number Replace System Fluids Replace System Fluids CH Cleaner 7 CH Cleaner Cleaner 29d23h 594837 Standby Standby StandbydA OCH it IEMENS CH Conditioner 6 CH Conditioner 29d23h -- IEMENS 9583727649 594837 MWDDNYYYY Capabilities SIEMENS Category Time Replace System Fluids Alert CH Cleaner 7 CH Cleaner CHWash CHwash wHWash 29d23h 11/19/2019 7483927482 7483901304 10495 265987 The CH has insufficient CH Replace the CH Diluent The CH Analyzer has insufficient CH Diluent. Replace the CH Diluent The CH Anatyzet has insufficient CH Replace the CH The CH Analyzer has insufficient CH Replace the CH Diluent Sequence Expiration Expiratio.n Inventory > Supplies Overvew 12/10/2018 6 CH Conditioner MMJDD/YYYY The CH Anatyzer stopped processing sarnples because a capability is In The CHAnatyzer stopped processing sarnples because a capability is In CH Diluent IMT Diluent A CH Diluent Exp red OdOh 712/15/2019 11/19/2019 71/15/2019 9765894652 System > Status, deterrrine which is down Operator System > Status, deterrnine which capabdity is down Review Operator System > Status System > Statis 12/102018 AM 12/10/2018 AM Event Log for events related to that and perform appropriate Event Log for events related to that and perform appropnate Healthineers ' corrective actions. Resume analyzer corrective actions Resume analyzer Sequence Expiratio.n Inventory > Supplies Overview 12/10/2018 1024:37 AM Replace A-LYTE multisensor 5 CH Wash MWJDDNYYYY MWDD/YYYY Connected to pcc Maintenance > The foUmving rnaintenance activity is overdue: IMT Dai* Cleaning. The rnaintenance activity is overdue: IMT Dai* Cleaning. 11/19/2019 Lot Sequence Expiratio.n 11/19/2019 4 CH Maintenance > The mnaintenance activity is overdue Replacing the Source Larnp. The maintenance activity is overdue Replacing the Source Larnp. AM IMT Volume Status OBS Expiredion Lot Expiration Expiration Sequence Number Cancel The rnaintenance activity is overdue: CH Dalkly Maintenance. 12/10/2018 Replace IMT Replace std B salt Bridge B salt Bridge 29d23h 11/19/'2019 165987659213 365987659213 Patent Sample Processing Patient Sample Processtng The rnaintenance activity is overdue: CH Weekly Maintenance, The following rnaintenance activity is overdue: CH Daikly Maintenance. IMT StandardA StandardA 100% 0K 29d23h 712/15/2019 71/15/2019 11/15/2019 S'21t/2019 165987659203 365987659213 165987659213 265987459213 46895 23895 The rnaintenance activity is overdue: Checking the Larnp Coolant The rnaintenance activity is overdue. Checking the Lamp Coolant 11/262018 3:48:14 PM 11/262018 3:48: 14 PM 11/2612018 3:48:14 PM The foUlovhng rnaintenance activity is overdue: Replacing the Dilution Ring The following rnaintenance activity is overdue: Replacing the Dilution Ring The foUlovhng rnaintenance activity is overdue: Replacing the CH Reaction Ring IMT Diluent Diluent 100% 0K OK 52/19/2019 11/19/2019 265987 10495 Cuvette Segments. The following rnaintenance activity is overdue: the CH Reaction The mnaintenance activity is overdue: Replacing the Source L.amnp, A Sensor PM IMT Sensor INT Sensor Exp red Expired OdOh 11/152018 265987 R•ng Cuvette Segr•nents Ring Cuvette Segments PE PEP Monday, December 10, 2018 11:31:14 AM Morday, December 10, 2018 Morday, December 10, 2018 AM Morday, December 10, 2018 11:31:14 AM Monday, December 10, 2018 Monday, December 10, 2018 AM Monday. December 10.2018 11:37:31 AM Monday. December 10, 2018 AM Monday, December 10.2018 11:37:31 AM Monday, December 10.2018 AM