Replacing the IMT Fluids on the Atellica® CH Analyzer with Sample Handler Video Version 1.25 and Below
This video provides instructions on how to perform the task of replacing the IMT fluids on the Atellica® CH Analyzer with the Sample Handler.
This video will show you the steps on how to change the IMT fluids on the Atellica Chemistry analyzer with the sample handler, the empty fluids are located in the front of the chemistry analyzer on the top left to replace the empty fluids, first ensure that the analyzer is in ready processing or standby state on the module display. The analyzer status is displayed in the upper right corner. Then lift the empty fluids lid located on the top front cover of the chemistry analyzer. Next on the module display, select IMT supplies. If this window displays with options to either perform I empty, advanced clean or replace. I empty supplies. It means that and IMT advanced clean is required before the next multisensor replacement. If you were only replacing the empty fluids and are not replacing the a light multisensor, select replace. I empty supplies and wait for the analyzer to pause. Locate and remove the appropriate I empty fluid container. The chemistry analyzer houses three EMT fluid bottles. A light I empty standard B plus salt bridge, a light I empty standard A. And a light I empty diluent discard the correct empty. I empty container as per laboratory protocol. Do not touch the cannulas in the key areas at the bottom of the system fluids drawer. The cannulas can puncture skin. Using the module display, scan the new I empty fluid bottle. Ensure the correct lot sequence and expiration display. Holding the new IMT fluid vertically with the septum facing down. Insert the new container into position. The empty fluid containers only fit in the designated position when the cannula punctures the septum. Close the I empty fluids lid. To complete the procedure, select OK on the module display.
module display,module state,system fluids,chemistry analyzer,analyzer,ensure,cannulas,imt fluids,sample handler,hereinafter collectively,reagent loader,imt supplies,operator manual,supplies,material,training,country,replacing,siemens,close,labmanager,note,software,functionalities,contact,trademarks
SIEMENS CH LabManager Please note that the learning material is for training purposes only! For the proper use of the software, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. SIEMENS IMT Supplies Supplies Replace IMT Supplies Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software CI-CH and hardware available at the time of the training. i IMT Fluid Container Scanned IMT Fluid Container Scanned System Fluids Module State IMT Supplies LabManager Standby The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Siemens Healthineers presents The IMT Advanced Clean maintenance procedure is due. 2 Standard A 2 IMT Standard A Note: Some functions shown in this material are optional and might not be part of your system. Lot 1222222222-22222 Expiration 5/5/2017 AM Expiration 5/5/2017 12:00:00 AM Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your Reagent Loader country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact Replacing Fluids To begin the Advanced Clean before replacing the A-LYTE mulitsensor, select Perform IMT Advanced Clean. your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be SIEMENS liable for damages. on the [email protected] CH Analyzer Healthineers Module State System Fluids Supplies IMT Supplies All names and data of patients, parameters and configuration dependent designations are fictional and examples only. To replace IMT supplies directly without performing the IMT Advanced Clean, select Replace IMT Supplies. To All rights, including rights created by patent grant or registration of a utility model or design, are reserved. with Sample Handler SIEMENS Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affliates. SIEMENS All other trademarks and brands are the property of their respective owners. To resume the IMT System, place the IMT Fluid in the holder and close To exit, select Cancel. the IMT Supplies lid. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright O Siemens Healthcare GmbH 2019 OK Cancel Perform IMT Advanced Clean Clean Replace IMT Supplies Reagent Loader Global Siemens Healthineers Headquarters Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84 0 siemens.com/healthineers