Replacing the Source Lamp on the Atellica® CH Analyzer Video Version 1.25 and Below
This video provides instructions on how to perform the as needed maintenance task of replacing the source lamp on the Atellica® CH Analyzer.
video will show you the steps on how to change the Ch analyzer source lamp. The Ch analyzer alerts for Lamp replacement based on lamp intensity. To change the lamp, the system must be set to diagnostic state. This can be done by selecting system in the command bar. Next select operator diagnostics. Select the chemistry module icon, select enter diagnostics, select yes. In subsystems, select lamp. Select replacing source Lamp select perform. Select yes, wait for the analyzer to prepare for maintenance. The covers unlock automatically if the back cover re locks before you were able to open it, select the button in the software to unlock the back cover. Open the back cover. Once the cover is opened, remove the bulb access plate. Next, press the side power button to the left of the lamp to turn it off. Do not touch the lamp housing. The lamp housing becomes hot with use and can cause burns. Allow the lamp to cool for approximately 5 minutes before touching any components. Disconnect the lamb connection by squeezing the sides and pulling apart. Do not look directly at the Ch UV light in the source lamp assembly or at its reflection from a shiny surface. The UV light can cause eye damage. Loosen the lab thumbscrews. Pull the lamp out and discard for laboratory protocol. On the new lamp, do not touch the glass portion of the lamp. This can cause damage to the lamp. If the lamp is dirty, clean it using lint free towelling moistened with ethanol. Insert the new lamp into the housing line up the pin on the lamp housing with the matching slot on the lamp base. Rotate the lamp into the housing until it slides into place. Reconnect the lamp connection. Tighten both thumb screws. Turn the lamp power on. Replace the bulb access plate. Close the back cover of the analyzer to exit diagnostic state. Select continue, select close, select exit diagnostics, select yes to complete the procedure. Select the maintenance tab in the Command bar, select schedule. Select the chemistry module from the dropdown menu. Select replacing source lamp in the software mark, replacing source lamp as completed. After one hour, perform an auto check, wait for auto check to complete before analyzing samples. Perform QC if QC fails, re calibrate the failed assays and repeat QC before processing patient samples.
command bar,lamp connection,lamp coolant,lamp housing,select continue,lamp,select,close,change,wait,analyzer,open,exit diagnostics,auto check,source lamnp,back cover,lamnp coolant,operator diagnostics,ch autocheck,add activity,source lamp,operator,activity,autocheck,module,system,scheduled,clean,perform,subsystems
Atellicae Solution Version 1 .14.0.2983003 Atellicae Solution Version 1 Atellica• Solution Version 220.127.116.1183003 Atellicae Solution Version 18.104.22.16883003 Atellicae Solution Version 1'14.0.2983003 SIEMENS SIEMENS . SIEMENS. Solution Version .22.214.171.12483003 SP00107i SP00107 SP00107• Healthineers SIEMENS .. SIEMENS 11 Healthineers Subsystems System Seno Sento Setto InveNory ln•æntory Invaltory Invertory Inventory Invertdory Inve•tory Inve-dory Orders Report Software Updates Operator Diagnostics Operator About SH SH 1M ATELLICA ATELLICACH ATELLICA.CH ATELLICA-CH Module: Frequency Time: All Performed By: AlI System VMM ATELUCA Module Activity Scheduled Sat-16 Mon-18 Tue-19 Sun-17 Wed-20 Wed -20 Fri-22 Subsystems Dragnostic Name Diagnostic Name Name Descripton Description All Mark as Completed Confirmation Standby Replacing the Source Lamnp Replacing the Source Lamnp? Replacing Source Lamp Replacing; the Source Lamnp: Replacin)g the Source Lamp Replacing CH Autocheck AutochecK CH Autocheck Source Lamp Perform Replaceng the Source Lamnp: All Time Inspecting the Washer Probes Checking the Lamp Coolant Checking the Lamnp Coolant Checking Lamnp Coolant Ch•ecking the Lamp Coolant Checking the Lam. Coolant Checking the Lamp Coolant Lamp Tme Lamp Confirmation All Performed By Capabilities CH Datkly Maintenance CH Dai,' Maintenance CH Dai/' Maintenance IMT Replacing the Source Lamnp? Replacing the Source Lamp Enter the activity name, Search Text 01/017/'2019 0:00 Replacing the Source Lamp: Scheduled by Operator Replacing the Ring Cuvette Segments Replacing CHe CH Reaction Ring Cuvette Segments 02/01/2019 co00 01/01/2019 coo0 Perform: Replacing the Source Lamp? Perform: the Source Lamp? Replacing the Source Lamnp: Replacing the Source Larnp? Scheduled Automatically Lamp Replacing the Dilueaction Ring Cuvette Segments Replacing the Dilution Ring Cuvette Segments 01/017/2019 0:00 Autocheck Autochec•k Scheduled by Service 2. After completing the procedure, select Continue. CH Maintenance 01101/2019 coo 01/01/2019 coo Connected to PCC Failed Yes No Inspecting and Cleaning the CH Probes and Mixer Impellers Overdue I nspect and Clean I r.spect and Clean Inspect Clean Inspect and Clean I nspect and I nspect and Ckan F"ATELLICA-CH Inprocess Inspecting and Cleaning the Probe Wash Stations Close Yes No Unlock Covers Covers Continue Completed Clearing the CH Fan Filter Cleaning CH Fan Filter Cleaning the CH Fan Filter 02/101/2019 7:00 01/017/2019 0:00 02.101/2019 7:00 Back Cover Front Cover Not Performed Not Perform... CH Weekly Maintenance Back Cover With Comments Checking the Larnp Coolant Checking the Lamnp Coolant 102/017/2019400 02/017/2019400 02/17/2019600 ATELLICA.CH ATELLICA-CH Replacing Ihe 'MT Peristaltic Pump Tubing Replacing the IMT Peristaltic Pump Tubing 15:03 Operator Diagnostics Exiter Diagnostics Exiter Diagnostics? Eniter Diagnostics Add pagnost»cs Exiter Diagnostics Operator Diagnostics Exit Diagnostics Exit Diagnostics? Add Activity.. Add Activity. Add vity axe PEP Thursday, 14, 2019 11:19:57 Thursday, February 14, 2019 Tuesday, 19, 2019 PM Tuesday, February 19, 2019 PM Tuesday, February 19. 2019 PM Tuesday, July 10, 2018 112730 AM LabManager LabM an ager Thursday, February 14, 2019 AM Thursday, 14, 2019 AM Thursday, February 14, 2019 11:1967 AM