Resolving Carrier Jams on the Atellica® Magline Transport Job Aid Version 1.25 and Below
This job aid describes how to resolve carrier jams on the Atellica® Magline Transport.
Atellica® Sample Handler SIEMENS Healthineers Resolving Carrier Jams on the Atellica Magline Transport Job Aid Issue: Carrier Jams on the Atellica Magline Transport Description: Sample carriers are stuck at a diverter, or they are not moving due to debris or obstruction on the track. Online Help Reference: Clearing Carrier Jams on the Atellica Magline Transport Troubleshooting Steps: 0011001234 527 1. Stop the Atellica Magline Transport on the System Status screen 2. Locate any debris causing the jam and remove 1 or more Atellica Magline Transport covers to access the debris and clear the jam 210914552A-211 3. Remove any debris and clean the area, if appropriate. To clean, use lint-free tissues or gauze to clean Carrier at a diverter position first with 10% bleach solution, then water, then dry 4. Slide the carriers apart manually to clear the jam condition NOTE: The Atellica Magline Transport and SH do not initialize • Warning: Do not remove carriers from the Atellica Magline Transport, the carriers have strong magnets in the if a carrier is present in the diverters. If appropriate, ensure base, and handling incorrectly may result in injury. all analyzer probes and SH / SHC grippers are off the Atellica 5. Ensure carriers are clear of the Atellica Magline Transport diverters Magline Transport and in a safe position before starting the system. This enables the free movement of the carriers when 6. Reinstall the Atellica Magline Transport covers the Atellica Magline Transport initializes. 7. Navigate to the System Status screen for the Atellica Magline Transport and select 1 8. If the problem persists, restart the PCC Template Effective Date: 27-MAY-2019 | HOOD05162003010284 Effective Date: 19-Jul-2020| HOOD05162003102693 9. If problem persists, contact the local technical support provider for assistance Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only. For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 2 siemens.com/healthineers Effective Date: 19-Jul-2020| HOOD05162003102693 Unrestricted © Siemens Healthineers, 2020