Resolving Definition Setup Issues on the Atellica® Solution Job Aid Version 1.25 and Below
This job aid describes how to resolve calibration, QC, and test definition setup errors on the Atellica® Solution.
Atellica® Solution SIEMENS Healthineers Resolving Definition Setup Errors Job Aid Add Calibrator Definition Calibrator Material Assay Type Material Name Material ID Lot ID Expiration Date Revision CH Tox Cal TO 7290999 5/31/2018 Active V Store Onboard Measuring Interval Verification Material OO Search Select Tests Assigned Values 6AM Atc_3 AAG Cal Code Assay Level 1 Level 2 Units Issue: Calibrator, QC & Test Definition setup AAT Acet Alb 102 Acet 9.2 mg/dl X AlbP ALP_2c ALT ALTPLC Amm Amp1k Description: Calibrator, QC and Test Definitions are set-up incorrectly Amp300 Amp500 Amylas APO A1 APO B ASO_2 AST ASTPLC B2M Online Help References: About Installing Calibrator Definitions / About QC BLUE Bnz200 Bnz300 Brb200 Brb300 Bupn Definitions C3 C4 Ce Caff Carb Che Troubleshooting Steps: ? Add Assay by Cal Code Print Save Cancel • Check the Operator Event Log for events associated with QC definitions, Add New QC Definition X and resolve those events Control Details Control Name LYPH CHL1 Control Lot 14491 Specimen Serum/Plasma The Auto Calibration with QC feature requires both the setting in Setup- Control ID 14491 Control Lot Expiration 12/31/2019 15 Control Type Routine • Level 1 Jsage Primary Test Definition – Calibration and the setting in the QC > QC Definition Store Onboard Perform QC with Calibration Perform QC on Pack Change window for the appropriate control material to be enabled. Ensure that Tests H est Name Units SD Low/ High/ Mean / +/- Open Stability Open Stability Range Result the settings in both areas are enabled Days Hours Clear All Ranges L a Alb g/dl 4.34 4.07 10.27 Observed Ranges • Issue Scanning: Make sure the software is on appropriate screen. Try a new K NA AAG AAT package insert from a different box. Insert sheets can also be found on Acet AFP Document Library. aHAVM Alb AIbP ALP 2c • Editing QC Definitions: When removing a method from the QC Definition, ALT ALTPLC Amik AMY_2 you must remove the method from any panels and from the QC scheduler. ? Save Cancel NOTE: CH Test Definition changes are included in software updates. 1 IM Test Definitions are updated every time the Master Curve Test Definition barcode is Template Effective Date: 27-MAY-2019 | HOOD05162003010284 Effective Date: 8-AUG-2020 | HOOD05162003112876 scanned in for a new IM reagent lot. Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only. For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 2 siemens.com/healthineers Effective Date: 8-AUG-2020 | HOOD05162003112876 Unrestricted © Siemens Healthineers, 2020