Resolving Waste Subsystem Issues on the Atellica® IM Analyzer Job Aid Version 1.25 and Below
This job aid describes how to resolve waste subsystem issues on the Atellica® IM Analyzer.
Atellica® IM Analyzer SIEMENS Healthineers Resolving Waste Subsystem Issues Job Aid Issue: Issue with the waste subsystem Description: Waste vacuum out of range / waste bottles may or may not have liquid waste Online Help Reference: Troubleshooting System Issues > Issue with the waste subsystem Troubleshooting Steps: 1. Shutdown and restart the IM analyzer: A. Navigate to System > Status > analyzer icon and select B. If no tests processing, select Yes C. If tests processing, select the appropriate shutdown option D. Wait for the analyzer status to display "Not connected to PCC.“ E. Ensure the software is shutdown as indicated by the computer’s green power light no longer being visible NOTE: If the IM analyzer power is off: For > 30 minutes, unload, manually mix, and reload all onboard reagents F. Turn off the power switch to the analyzer For > 2 hours, discard any onboard reagents G. Wait 30 seconds before restarting the analyzer H. To restart the analyzer, power on the analyzer and wait for the analyzer’s module display to come up. Ensure the analyzer is in Ready or Standby State 1 2. If problem persists, contact the local technical support provider for assistance Template Effective Date: 27-MAY-2019 | HOOD05162003010284 Effective Date: 11-OCT-2020| HOOD05162003129475 Unrestricted © Siemens Healthineers, 2020 SIEMENS Healthineers Please note that the learning material is for training purposes only. For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Atellica, and all associated marks are trademarks of Siemens Healthcare Diagnostics Inc. or its affiliates. All other trademarks and brands are the property of their respective owners. Product availability varies by country and is subject to varying regulatory requirements. Please contact your local representative for availability. Copyright © Siemens Healthcare GmbH 2020 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 2 siemens.com/healthineers Effective Date: 11-OCT-2020| HOOD05162003129475 Unrestricted © Siemens Healthineers, 2020