System Messages and Message History (XA30)
This video provides an overview about the Message History and different System Messages.
System messages are displayed as icons in the status bar. In a single monitor setup, all system messages are summarized in one message history. In a dual monitor setup, the system messages on the main monitor refer to EMAR system related issues and on the second monitor they refer to Mr View and go related issues. These messages are generally classified as errors or information. In addition, there is an icon to open the scanner, an acquisition dialog box on the main monitor. This dialog box includes warnings related to Mr Hardware. For example, the cooling system, coil selection, and resonance frequency. Different icons in the status bar inform you about the number and type of messages. The message history icon is displayed. If you have no new or unread messages. If you have several messages in the message history, the icon with the highest priority is displayed in the status bar. Click on the icons in the status bar to open the message history and dialog box in order to read the details of the message, open the message. You should always check the icons in the status bar and open the related dialogs. If workflows are interrupted or errors are shown to continue with the workflow, read the details of the messages displayed in the message histories. Remember, if you open the message history dialog box, all messages will be automatically set to read and the icon will change on the screen. You should therefore ensure that you check all messages in your message history before you continue scanning.
status bar,message history,system messages,message,messages,displayed,icons,errors,icon,open,read,pt oblem,marked protocols,rneasurement system,detected reliably,autoabqn landmarks,operator manual,manual adjustment,resonance frequency,system,manual,frequency,hardware,autoahqn,persists,training,coverage,check,material,information
01:47 improve the coverage o' the AutoAhqn sccnlt. 01:47 MR Scanner & Acquisition Message History System messages @ Siemens Healthcare GmbH, 2020 Date For the proper use of the software or hardware, please always use the Operator Manual or Frequency not Frequency not converge 4/15/2020 PM 4/tS/2020 PM PM Preparation of rneasurement system has tailed. Preparation of rneasurement system has faded. rpaqrement system system The resonance frequency (Ould not be detetrmined. The resonance frequency could not be determined. Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. You can retry or restart the workplace You can or restart the workplace Frequency not converge This material is to be used as training material only and shall by no means substitute the Operator It the pt oblem still persists, please use the manual adjustmnent It the ptoblem please use the manual adjustment It the pt oblem still persists, please use the manual adJustment It the ptoblem still persists, please use the manual adJustment and message history plat'orrn 'or analysis. plat'orrn 'or 'or Manual. Any material used in this training will not be updated on a regular basis and does not Adjustment did not converge. MRI ADS 4/tS/2020 PM AutoAhqn Some AutoAbqn could rwt be detected reOiabty. Please check the marked protocols or try to AutoAhqn Some AutoAbqn landmarks could rot be detected reliably. Please check the marked protocols or try to Laronarts not "e detected or fry AutoAbqm: Sorn,e AutoAbqn landmarks could bote detected reliably. Please check the marked protocols or try to necessarily reflect the latest version of the software and hardware available at the time of the Recerved too low MR Signal was too low. VR was too low improve the anatorn.cal coverage o' the AutoAhqn Improve the anatomical coveraqe of the AutoAhqn scout. improve the coverage o' the AutoAhqn improve the anatomical coverage ot the AutoAnq scout. improve the coverage of the AutoAhqn Clicking on this icon provides access Piease pat.ent posit.on.ro. and adjustrr*nt Please check patient (0.1 selection and ad'ustrnent AutoAI'-gn Some AutoAngn lardmarts training. Go Cancel to recent user messages. volume, coubd not be detected reliably the p' oblem persists. please use the manual problem persists. '*ease use the manual adjustrrrnt Please check marked protocols or The Operator Manual shall be used as your main reference, in particular for relevant safety platform further analysts. try to imprm•e the anatomical information like warnings and cautions. coverage of the AutoAlgn scout. the the •s Note: Some functions shown in this material are optional and might not be part of your system. The Close the door It the problem *lease contact Sr-mens Semvice. It the problem *ase contact Sr-mens Semce. information in this material contains general technical descriptions of specifications and options as Show only rtess."es Show onw The reproduction, transmission or distribution of this training or its contents is not permitted without well as standard and optional features that do not always have to be present in individual cases. express written authority. Offenders will be liable for damages. Stimu Certain products, product related claims or functionalities described in the material (hereinafter All names and data of patients, parameters and configuration dependent designations are fictional collectively "Functionality") may not (yet) be commercially available in your country. Due to and examples only. regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current All rights, including rights created by patent grant or registration of a utility model or design, are information. reserved. Stimu