Ultra Flex 18 Large (Numaris X)
This video explains the handling and features of the UltraFlex 19 Large Coil when examining the shoulder.
The Ultraflex 18 large coil can be used for examinations of different body regions. In this video you will learn how to use the coil for examining the shoulder. A general hint to start with. Position the flat side of the coil against the region of interest. To examine the shoulder with the Ultraflex large 18 attached this strap to the opposite side of the patient table of the shoulder to be examined. Position the coil. Ensure that the region to be examined is contained within the side flaps of the coil. You can easily plug in the coil cable with the slide connect mechanism. You will see a message that the coil is correctly plugged in on the PD. Ensure that the coil cable does not hang over the sides of the patient table. Secure the coil with the strap. Additional sandbags can be used to help secure and keep the coil in position during the exam hints. To examine the hip position the patient feet first and follow the steps for patient positioning similar to the shoulder for examinations of the knee or the foot or ankle. Position the coil in order to ramp the part of the body you want to examine. Use appropriate positioning AIDS to ensure patient
patient table,coil cable,body,ensure,coil,position,examinations,shoulder,region,examine,strap,examined,secure,operator manual,hereinafter collectively,siemens healthineers,optional,system,information,magnetom,country,training,material,siemens,healthineers,learn,part,note,ultraflex,software
Healthineers SIEMENS Healthineers SIEMENS Healthineers Healthineers SIEMENS Healthineers No patient registered @ Siemens Healthcare GmbH, 2018 MAGNETOM systems Please note that the Learning material is for training purposes only! ULtraFLex Home For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training spine 32 18 Large will not be updated on a regular basis and does not necessarily reflect the latest version of the software and UltraFlexLarge 18 hardware available at the time of the training. The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. COIL PLUGGED Note: Some functions shown in this material are optional and might not be part of your system. The information MAGNETOM in this material contains general technical descriptions of specifications and options as well as standard and The reproduction, transmission or distribution of this training or its contents is not permitted without express optional features that do not always have to be present in individual cases. written authority. Offenders will be Liable for damages. UltraFlexLarge 18 A BioMatrix System AIL nomes and dato Of patients, parameters and configuration dependent designations are fictional and Certain products, product related claims of functionalities described in the material (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, the examples only. future availability of said Functionalities in any specific country is not guaranteed. Plaese contact your local Siemens Healthineers sales representative for the most current information. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. 2)