YSIO X.pree Restore Collimation
This job aid will cover the steps to follow to Restore Collimation on the system.
YSIO X.pree: Restore Collimation XPRF The “Restore Collimation” icon can be found on the touch user interface (TUI) of the X-Ray tube. YSIO X.pree = = Emergency_3396cede Examination Positioning Clinical protocol step name Acquisition Type Patient Size M Wall Stand Al Measuring Field -- kV 121 + Recall Collimation Show more 21 180 0m mmCx 00 | [1 17.0 00x 15.8 09 | Mamal Align tube to Wallstand Detector so that X-ray won't V go beyond Detector at any side Note: When the “Recall Collimation” icon is selected, the system recalls the last collimation size used for an exposure. Davies, William Chest p.a. L Abdomen a.p. upright Abdomen a.p. supine Acquisition Type Table M Patient Size - KV 81 Dose Measuring Field 2 + + - Recall Collimation Show more + Centered lllll] Current: fo115 Center Top Align III) Desired: fo115 8| 115 cm Distance mmCU: 0.0 | | | 11 42,4 cm x 34,8 cm | Automatic (ACSS) V Error! Unknown document property name. Error! Unknown document property name. Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 SIEMENS Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Job Aid Effective Date: 09/07/2021 | HOOD051620032118151 Healthineers Job Aid | X.pree: Restore Collimation Effective Date: 09/07/2021 | HOOD051620032118151 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter co llectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and c autions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be comme rcially available in your country. Due to regulatory requirements, the future availability of said Func tionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2020 Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Phone: +49 9131 84-0 Error! Unknown document property name. Error! Unknown document property name. Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2020 SIEMENS Master Template Effective Date: 3/5/2020 | HOOD05162003074494 Job Aid Effective Date: XX/XX/XXXX | HOOD/HILS Healthineers Page 2 of 2