Cleaning the Rinse Stations Video
This video provides instructions on how to properly clean the rinse stations on the ADVIA Centaur® XP system.
Turn system mechanics off and lift the instrument lid. Remov the reagent probes cover. Push the reagent probe assemblies tored the incubation ring by pushing the Motors. Use a disposable transfer pipette to add 0.5 to 1.0 milliliters of cleaning solution to the rinse station. Clean each rent station using the small maintenance brush. Gently insert the brush into the rent station to a depth of 2.5 inches and swirl the brush. Remove the brush. And then remove the cleaning solution with a transfer pipette. Clean each rent station by wiping with lint free tissues or gauze saturated with cleaning solution. Wipe the rent station with lint free tissues or gauze saturated with the ionized water. Close the top cover and turn system mechanics on.
- as needed maintenance