SOMATOM go. platform Changing the API of One Scan Range Video
This video provides instructions on how to change the API of one scan range when operating a SOMATOM go. platform scanner. Disclaimer: Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. Functionalities described in the material or parts of this functionality may not yet be released for customers and not yet be commercially available in every country. Due to regulatory requirements, the future availability of said functionalities or parts thereof in any specific country is not guaranteed. The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients and institutions that are used in these instructions for use are entirely fictional. Any resemblance to names of existing people or organizations past or present is entirely coincidental. All parameters and images shown in this online training are examples. Only the parameters displayed by your system are definite. Configuration-dependent designations, such as names of drives, network nodes, and databases, that are used in these instructions are usually not the same as the designations to be found on a particular installation of the system in a clinical environment. All rights, including rights created by patent grant or registration of a utility model or design, are reserved.
The CT system provides prerecorded automatic patient instructions consisting of three pre recorded instructions including expiration inspiration. Do not swallow, and in addition, specific instructions for cardiac and perfusion. This movie will show you the steps for how to change an automatic patient instruction for one scan range to begin, navigate to the timeline and hover over each scan range to display the audio control window and view the current patient instruction for that scan range. Notice the patient instruction for both scan ranges is inspiration for this simulation. We will change the patient instructions for the long low dose scan to expiration. Next, right click the highlighted area in front of the scan range you want to change the patient instruction for. Select delete to delete the patient instruction for that scan range. Right click the scan range. A context menu opens from the context menu, select Open Edit mode. The Edit Mode toolbar will display above the timeline. Navigate to the audio control area and deselect apply to all. Select the patient instruction dropdown. And select expiration. Select the command icon and drag it to the highlighted area in front of the scan range in the timeline. The patient instruction has now been changed to expiration. Congratulations, you have completed this exercise.
SIEMENS Thorax (Adult) Recon Ranges Congratulations. You have completed this exercise • Inspration • Inspiration Audio Control Exam Contrast Tracking Test eee Mode 5.3 8.1
- somatom go.
- somatom go
- somatom go. up
- somatom go.now
- somatom go. top
- somatom go. all