epoc® Blood Analysis System with NXS Host Performing a Whole Blood Test Video
This video will show you the steps on how to perform a whole blood test on the epoc® Blood Analysis System with NXS Host.
This video will introduce you to the steps on how to perform a whole blood test on the epoch blood analysis system with NXT host. To perform a whole blood test, you will need the following supplies. The epoch reader. The epoch next host. A new epoch test card and a syringe. Please wear the proper personal protective equipment when handling any patient blood specimens. To start, press the power switch to turn on the epoch reader. The power LED will turn green, indicating the epoch reader is on and ready for use. Next, turn on the host by pressing the power button located on the upper right side of the host. Sign in to the epoch host using your sign in User ID and password. Or by scanning your user ID barcode. When the host connects to a selected reader, the test status indicator will turn solid green. This means the reader is ready for use. Next, the host will display a test screen. The screen will then show insert new test card, select a test card and remove it from the pouch. The test card pouch should only be opened when you are ready to conduct a blood or QA test to assure a low humidity environment for the test card. Be sure to not touch the sensor modules contact surface or the blood sample entry port. Insert the test card into the reader immediately to avoid contamination. Do not place the test card on any surface before running the test. Hold the test card facing the sample port upwards. Insert the test card into the card slot of the reader by firmly pushing it in using a smooth single motion. Ensure the test card label shown here is facing down and the sensor module is facing towards the reader to be inserted first before inserting into the card slot. This will enable the barcode on the card label to be read correctly by the epoch reader. Be sure to lock the card into place once the card is locked into place, the reader will beep once. The host screen will then display the card lot number. And the test status indicator light will start flashing green. Next, the host screen will show that the system is calibrating. Do not inject the sample into the card until calibration is complete. The calibration process takes approximately 165 seconds to complete. Once calibration is complete, the system will beep twice. The host will display an inject sample message and countdown timer on the screen. Next, inject the sample using a syringe into the port on the test card. Hold the syringe barrel vertically between fingertips and thumb and perpendicular to the test card. Do not introduce air bubbles into a collection device. If air bubbles are present, remove immediately after the sample is collected. Using slight downward pressure, secure the syringes lure tip into the blood sample entry port of the test card. Rotate the syringe up to 1/4 turn to ensure a good seal while maintaining downward pressure. Use the index finger of the other hand to steadily depress the syringe plunger with a single smooth, continuous motion until prompted to stop. The reader will beep and the test status indicator will flash green. If the sample is injected into the card too soon. Or too late. An error message will appear and the system will abort the test. Remove the syringe from the test card by tilting it as opposed to straight up to ensure sample integrity. While the sample is being analyzed, enter the patient ID into the host. Once the information has been entered into the host, a check mark will be visible on the blue circle and the patient ID is shown in the action bar. Patient ID can also be viewed in the edit test details section of the software. Answer yes or no to the hemodilution question and press confirm to proceed with the blood test. Next, choose select test in the software. Select the test panel from the list, select the proper buttons to customize the test options. Select confirm. Once all information has been entered. Other patient information, such as patient age, gender and height can be entered into the edit test details section. If temperature correction is needed, the information can be entered into this area as well. Once complete, select confirm in the software. The host will display analyzing sample when the test is in process. When the analysis is complete, the results will display on the host screen. If any data is missing, the missing required data message will appear on the screen, prompting the operator to enter the missing data. In order to view the test results. Remove the test card once the results are displayed on the host screen, discard according to laboratory protocol. Select the save and close button to save the test data and close the test window. The X in the top left corner can also be selected to closeout. The test data window. This completes the video on how to perform a whole blood test on the epoch blood analysis system with NXT host.
2020-05-29 50012187-02 510747 00 12 123 1234 7123 2021-12-24 52 11-21190-40 606740 2123 100% 7.35 10.3 97.4 13.3 -2.2 99.9 128 4.5 1.05 13.6 20 epoc® Blood Analysis System with NXS Host Performing A Whole Blood Test Video epoc BLOOD ANALYSIS CE BLOOD AN REF 10736515 IVD pH Na+ K+ p02 Hct LOT 10-19347-20 SN 510747 Epocal Inc. pC02 Lac Ca++ Crea LOT SN Ottawa, ON Canada siemens.com/epoc 11-Aug-21 14:10 Version 4.6.4 About User ID SIGN IN 1 2 ABC 56 7 8 9 9 0 c v bnm Regional Bryan Degala RADIOLOGY 11-Aug-21 14:16 a s Connecting. Do not insert card. You can edit Test Details now. Edit Test Details Patient ID * Required to view results Scan or enter Patient ID Select Hemodilution * 2 3 4 Configuring Rdr26415. Do not insert card 3 4 card. 10s remaining. X Blood Test: Rdr26415 card. 1s remaining. qwert a sd n m Insert new Test Card. Show me how hide Glu BUN LOT 11-21190-40 SN 606740 Crea 11-Aug-21 14:35 Show less Reader: Rdr26415 78% battery Test Card inserted. Select Sample Type Enter patient information Patient temperature, age, gender, height Scan or enter Patlent ID f g h v b Test Card inserted Calibrating. Do not inject sample 02:52 remaining qwerty f g wertyu qwertyu 11-Aug-21 14:58 00:02 remaining h j Inject Sample 07:29 remaining e r q w a sdf g on 11-Aug-21 15:56 Analyzing sample 00:31 remaining qwertyuio 2 3 yuio 12 3 4 5 6 78 9 0 n # $ % & - + ( ) Next 456 7 89 0 Patient 1234 Select tests Yes No Required fields completed Enter sample information mple Type Venous Mixed-venous Capillary Cord Cord Arterial Cord Venous Unknown Arterial Clear O Arterial Draw site, Allen's test Enter respiratory information Enter Comments Comments Gases+ temperature, age, gender, height mple information Allen's test Electrolytes eMP Custom Selection pH Chem+ Ca++ TC02 Meta+ 00:10 remaining Cre Enter sample Delivery system, Mode, FiO2, Flow, RQ, .. X Patient 1234 11-Aug-21 15:57 00:04 remaining Calculating Results. 9-Apr-21 09:42 CLOSE & REPORT REJECT TRANSMIT CRITICAL RESULTS TEST mmHg mmol/L BE(ecf) 11-Aug-21 15:47 SAVE & CLOSE RESULTS cHCO3 iQC cnc BE(b) iOC show cHgb Close test? NO YES Run Blood Test View Blood Test History QA Testing Settings o p as d w e Please note that the learning material is for training purposes only. For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively "Operator Manual") issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Please note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively "Functionality") may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Siemens Healthcare GmbH 2022 SIEMENS Hea Healthineers