SOMATOM go. UI Navigation for Physicists Video
SOMATOM go. UI Navigation for Physicists Video
The introduction of the new Somatom Go systems. There is a new software interface that may be unfamiliar to most with strict quality control regulations in place by hospital, state and country. Physics testing on these systems has become an annual requirement. These tests include kedd dose measurement testing that requires specific protocols to be used. The purpose of this video is to show physicists how to access hidden dedicated testing protocols for KDDI measurement testing on Somoto Mgo systems. The steps shown in this video include how to unhide ctdi measurement protocols for both adult and pediatric body and head modes. Additional steps included in this video are how to register a patient. Choosing the see TD protocols and selecting the various parameter settings for protocol modification purposes. The process begins by logging into the system as a superuser. This is accomplished by selecting superuser from the list of users located in the top right corner. When the syngo setae login screen appears, enter the password and select the OK button. From the scheduler window, click on the settings icon in the top right corner. This may also be known as the cogwheel from the list of options select exam designer. From the select Protocols Tab, select the option Siemens protocols and deselect customer protocols. This will reduce the overall number of protocols visible, so it is easier to identify the semans default protocols. Select the specials folders for both adult and child protocols. Check the boxes for the skeddy measurement protocols under both adult and child headings. Select unhide in the bottom left corner of the screen. This will display the protocols in the examination card. Close the window by selecting the X in the upper right corner. Click examination, then select register patient in the top right corner. In the patient registration tab, enter test data for the last name, patient ID, and date of birth. Click exam, the message application is starting. Please wait will display. When the examination card opens, the protocol selection tab displays. Select the unhidden Skeddy measurement protocols for adult and child by clicking on the specials folder. Select the protocol on the right. In this example, we have selected the body protocol. Click the Add icon to register the protocol. Click OK to Ally the protocol. Right mouse click on the scan timeline and select show parameters from the list of options. This will display the scan parameters so they can be verified or modified. Click on the Orange Recon Square in the top right corner to display the Recon parameters. To make changes to the reconstruction parameters, click on the desired Recon tab and make any necessary changes. Select confirm to load the protocol and then go to activate the top of Gram scan mode. You are now ready to acquire the scan for testing.
administrator Scan S Recon 3 Scan S Scan S = 3 Scan S = Recon 3 Scan — Recon 3 Recon 3 Configuration Pa Configuration Panel Scheduler Scheduler superuser Scheduler. Administration Portal CTDI measurement... Protocol Parameters Recon Ranges Recon Ref. Protocols Confirm fi Click on the Settings icon Administrator Clinical Configuration 1firm Changes Patient Information Medical Information Physiological Signal Selecte Delecte Siemens Protcx:ols (Il 7) other user Check up Custuner Protocols Custo«ner Protocols (2) Contrast Protocol all Customner Protocols (2) Customer Protocols Protocols (2) (aka cog wheel). ft' i Selecte Patients Patients 2 Patient Delecte Test Medical Information Exam Information Patient Information Medical Information Daily QA 2s Siemens Protocols Customer Protocols Make tab Calibration Specials (2) Abdomen (0) Specials (2) Head (0) Head (O) Head (5) Abdomen (3) Xbdomen (7) Body Requested Procedure Medical Information CTDI Ernergency Register Requested re Resume CTDI CTOI CTDI Head Head Abdomen SIEMENS Un-hide CT DI measurement protocols & de-select Patient Patient Patient Type Patient Healthineers -IndicatK:ons: ACR -Indications: ACR ACR -Iruxatms: ror ACR neM ACR .Topogram „Topogram v'Topogram 'cular Thorax (2) Neck (0) Then select Exam Designer. grain Multi Phase CTDI Phase Phase Length I Wete syngo Remote Assist Delete I )elete Contrast Contrast protocol Customer Protocols Custuner Protocols Contrast protocol all protocols Alerts Vascular Neck (0) Ear Non Conti Body SPR Siemens Protocols Custuner Protocols Customer Protocols Custoyqt ProtoceE Patient Type Patient Patient Patients 2 syngovia all syngovia syngovta all all Screen recorder 0.3 Shoulder (0) Shoulder (O) Sinus 10 Middle Copy All Recon Parameters Registering a patient Pelvis (3) Contrast Protocol al' Contrast Protocol all orbit syngo CT syngovia Stones DE orbit SIEMENS SaveLog Savel_og CTDI Copy Recon Volume Parameters 007 Neck (0) Vascular Thorax (0) Vascular Cardiac (O) Dental Cohn Dose Dose Thorax (O) Vascular Abdomen (O) syngovta al' syngovta syngovta all Protocol Parameters various parameters. • Date of u,wyyyy ACR txny ACR rvice .Topogram "aste Rean Volume Parameters PelVß Bone Pelvis Bone 2020 Body Body Choosing the CTDI protocols Neck (1) CTDI measurement protocols for Exam Designer Vascular Abdomen (O) Hip Abdomen (O) Vascular Extremities (O) Pelv•s Bone (3) Neck Institution Spine (6) Spine (4) Pelwis Pelvis both Adult & Child. Pelvis (O) Pelvis (3) • Patient ID Shoulder (2) Selecting parameter settings Pelvis Bone Pelvis (3) Scan Favorites I Genera I 12 13 1' Auto Tasking I I Cmtrast Genera General Physio Recon Recon I — Recon 3 Auto Tasking Physio Scan I Physio Scan Vascular Cardiac (O) Cxalibration Examination System Check Lock 2s Trauma (O) Spine (O) Spine (0) RTP (0) CARE Dose4D Inline Scan Image Recon Ref. Recon Ranges Recon Ret. Image Cycle Acquisition Slice Thickrwss Inaenænt CARE Dose4D CARE Slice Thickness Body Part Absolute Stan Breath Hold Increment Transfer Width Height Center X Dose Notificabon Dose Notification Acq Dose N obfication Y HD Tawe Feed AbsoMe Start Delay Allergies Axial Kernel SAFIRE FAST iMAR kV Irnage Order Local Workflow 26 29 30 CARE W CARE w Time Imml CARE kv Irnml DLP T. or File Browser Thorax (3) T. Vascular Extremities (O) Specials (O) Lower Extremities (O) Select the Unhide button to Enter test data for Last Name, Patient ID, SOO Br40 SOO 100 Head to Feet Head (5) None ABDOMEN Rou ul Square Head (5) Head to Soft None Off Off Head to Feet Off Head to Feet Adult None Head to Head (O) u omen Thorax (0) Int—on lnt"ion Int—ention Select the "unhidden" CTDI protocols by Thorax IJpper Extremities (2) Vascular Extremities (O) upper Extremities (2) UJpper Extremities (O) Perform these steps for both Adult Lung Dose Lung LOW Dose Click Examination to begin Baby (O) and DOB then click the Exam button. selecting the Specials folder on either and Hand Wrist Height par parameters Lung LOW Lung Dose either the Head or Body protocol. and Child protocols. click OK Intervention (O) Activate Topogram scan Ccmfirm Ccmfirm patErut position Ccmfirm patErut Ccmfirm patEnt Sekct protcxol Sekct protrxol Appty protocol Appty protocdl Contrast protocol AppW protocd Vascular Head (O) Intervention (0) registering a patient. Examination Card. Click on the orange Recon square to Click each tab to view or modify the the Adult or Child tab. and select Show Parameters Enter Superuser password. Edit Mode display all of the Recon Parameters. Cleate Protocol Parameters various parameters. from the pop-up menu. Select the Specials folder for both to and and 2s 5s 2s Upper Extremities (O) Delete Delecte Selecte System Check Lock Adult & Child protocols. and Child protocols. Del ele Delete Stu 9:57 10:00 10:52
- SOMATOM go. UI