SOMATOM go.Now / go.Up / go.All / go.Top Air Filter Exchange - USA
SOMATOM go.Now / go.Up / go.All / go.Top Air Filter Exchange Job Aid - USA
Air Filter Exchange Job Aid SOMATOM go.Now & SOMATOM go.Up SOMATOM go.All & SOMATOM go.Top If the air filter is dirty, it might lead to the system overheating. In 5 easy steps, you can exchange the filter pads on your own quickly and easily without tools. There are 2 air filters at the top left and bottom right of the gantry back. 1 1 1 Air filter Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 SIEMENS Master Template Effective Date: 08/22/2017 | HOOD05162002647849 Healthineers Job Aid Effective Date: 02/13/2020 | US - HOOD05162003062237 Job Aid | Air Filter Exchange Effective Date: 11/30/2019 | HOOD05162003026134 1. Open the filter door (the door of the filter is closed with magnets). 2. Remove the old filter pad. 1 O 1 Filter pad Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 Page 2 of 4 Job Aid | Air Filter Exchange Effective Date: 11/30/2019 | HOOD05162003026134 3. Clean the grid, for example with a vacuum cleaner. 1 O 1 Filter grid 4. Place a new filter pad on the grid. 5. Close the air filter door. Please note: • It is forbidden to use the gantry without replacing the filter. • You will receive a notification when it is time to change the air filter. For information regarding the purchase of the filter pads, please contact your Customer Care Center. • If the filter pad is not very dirty, cleaning it with a vacuum cleaner is sufficient. Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 Page 3 of 4 Job Aid | Air Filter Exchange Effective Date: 11/30/2019 | HOOD05162003026134 Please note that the learning material is for training purposes only! For the proper use of the software or hardware, please always use the Operator Manual or Instructions for Use (hereinafter collectively “Operator Manual”) issued by Siemens Healthineers. This material is to be used as training material only and shall by no means substitute the Operator Manual. Any material used in this training will not be updated on a regular basis and does not necessarily reflect the latest version of the software and hardware available at the time of the training. The Operator's Manual shall be used as your main reference, in particular for relevant safety information like warnings and cautions. Note: Some functions shown in this material are optional and might not be part of your system. Certain products, product related claims or functionalities (hereinafter collectively “Functionality”) may not (yet) be commercially available in your country. Due to regulatory requirements, the future availability of said Functionalities in any specific country is not guaranteed. Please contact your local Siemens Healthineers sales representative for the most current information. The reproduction, transmission or distribution of this training or its contents is not permitted without express written authority. Offenders will be liable for damages. All names and data of patients, parameters and configuration dependent designations are fictional and examples only. All rights, including rights created by patent grant or registration of a utility model or design, are reserved. Copyright © Siemens Healthcare GmbH 2019 ………………………………………………………… Siemens Healthineers Headquarters Siemens Healthcare GmbH Henkestr. 127 91052 Erlangen, Germany Telephone: +49 9131 84-0 siemens.com/healthineers Published by Siemens Healthcare GmbH © Siemens Healthcare GmbH, 2019 Page 4 of 4
- somatom go.
- air filter